Physics Help Forum End correction in meter bridge practical

 Physics Physics Forum - General Physics Discussion and Physics News

 Apr 28th 2017, 11:59 PM #1 Junior Member   Join Date: Apr 2017 Posts: 2 End correction in meter bridge practical Could end correction be minus ? In meter bridge practical , I got minus end correction values .. is it possible ? Can anyone explain me what that means ? Thanks in advance .
 Apr 29th 2017, 05:52 AM #2 Senior Member   Join Date: Apr 2015 Location: Somerset, England Posts: 534 Hello Madhu and welcome. It is really good to see some practical work being done, this is just so important. The answer to your question lies in understanding where the end corrections come from. I have drawn a greatly exaggerated error to show the issue. When you read the bridge you measure from the balance point to what you hope are the ends of the slidewire. The ends of the slidewire are the points where the heavier copper bus bars are attached (eg soldered) on. They may not be exactly 1 metre apart. I have labelled these points M and N. Also the ends of the ruler may not be exactly aligned with these points. So I have shown two different errors, alpha and beta, one where the ruler sticks out and the other where the wire sticks out further. Measurements L1 and L2 are shown and the true lengths for the measurements are L1+a and L2+b. Of course a is positive and b is negative in this example. Does this help? Attached Thumbnails
 Apr 29th 2017, 12:45 PM #3 Junior Member   Join Date: Apr 2017 Posts: 2 Thank you. Thank you very much .. it was really helpful . I couldn't understand the reason for the minus end correction . Because of your explanation , now I understand .
Apr 29th 2017, 01:18 PM   #4
Senior Member

Join Date: Apr 2015
Location: Somerset, England
Posts: 534
 Originally Posted by MadhuNissanka Thank you very much .. it was really helpful . I couldn't understand the reason for the minus end correction . Because of your explanation , now I understand .
There is another possible error source, called connection or contact resistance.

Although the bus bars are heavy enough to consider them as having nelgigible (zero) resistance, the solder joint where they conect to the wire may not.

If each joint has a resistance r, and the test resistances are P and Q then measuring from the balance point each arm now sees a resistance of

P + r and Q + r

If we write Q/p = n or Q = nP then we have the ratio of the resistances as

(P + r) / (nP + r) instead of P/Q or P/nP

 Tags bridge, correction, end, meter, practical

,

,

,

,

,

,

,

,

,

,

,

,

,

,

# end correction in metre bridge

Click on a term to search for related topics.
 Thread Tools Display Modes Linear Mode

 Similar Physics Forum Discussions Thread Thread Starter Forum Replies Last Post Roma Electricity and Magnetism 1 Sep 15th 2014 04:53 AM Roma General Physics 0 Sep 13th 2014 08:40 PM hana1 Light and Optics 5 Jan 23rd 2011 11:03 AM bradycat Equilibrium and Elasticity 3 Nov 21st 2010 11:00 AM hana1 Light and Optics 10 Jul 3rd 2009 04:51 AM